Cross-calibration of the Siemens mMR: easily acquired accurate PET phantom measurements, long-term stability and reproducibility
Research output: Contribution to journal › Journal article › Research › peer-review
- Cross-calibration of the Siemens mMR: easily acquired accurate PET phantom measurements, long-term stability and reproducibility
Final published version, 2.24 MB, PDF document
BACKGROUND: We present a quick and easy method to perform quantitatively accurate PET scans of typical water-filled PET plastic shell phantoms on the Siemens Biograph mMR PET/MR system. We perform regular cross-calibrations (Xcal) of our PET systems, including the PET/MR, using a Siemens mCT water phantom.
LONG-TERM STABILITY: The mMR calibration stability was evaluated over a 3-year period where 54 cross-calibrations were acquired, showing that the mMR on average underestimated the concentration by 16 %, consistently due to the use of MR-based μ-maps. The mMR produced the narrowest calibration ratio range with the lowest standard deviation, implying it is the most stable of the six systems in the study over a 3-year period. MMR ACCURACY WITH PREDEFINED μ-MAPS: With the latest mMR software version, VB20P, it is possible to utilize predefined phantom μ-maps. We evaluated both the system-integrated, predefined μ-map of the long mMR water phantom and our own user-defined CT-based μ-map of the mCT water phantom, which is used for cross-calibration. For seven scans, which were reconstructed with correctly segmented μ-maps, the mMR produced cross-calibration ratios of 1.00-1.02, well within the acceptance range [0.95-1.05], showing high accuracy.
CONCLUSIONS: The mMR is the most stable PET system in this study, and the mean underestimation is no longer an issue with the easily accessible μ-map, which resulted in correct cross-calibration ratios in all seven tests. We will share the user-defined μ-map of the mCT phantom and the protocol with interested mMR users.
|Journal||E J N M M I Physics|
|Number of pages||8|
|Publication status||Published - Dec 2016|
Number of downloads are based on statistics from Google Scholar and www.ku.dk